The constant temperature and humidity display cabinet is required to be placed in a normal environment. The indoor temperature is within 16 ° C-26 ° C \ humidity within 30%RH-70%RH. 2.
Different power supply 220ⅴ or 380ⅴ based on different models, different, different, different, different, different, Please connect the power supply and view the plug logo to avoid damage. 3. Based on different models, use tap water sources or water tanks to replenish water replenishment.
If the water source replenishment method is adopted, please make sure that the connection valve should always be opened, otherwise the water replenishment cannot be replenished in time to damage the humidification equipment; if the water tank is used to hydrate, please replenish it in time to avoid damaging the humidification equipment damage. 4. The use of this machine should specify the operation of the person.
The unrelated personnel shall not press the switch on the panel at will, so as not to cause the automatic control program to be garbled, so that the whole machine cannot be used normally. 5. When adjusting the digital values of the measurement and control instrument, do not adjust the digital value to 00.
0 or 0.00. So as not to be chaotic, affect normal use.
6. If the failure or the machine work is abnormal, please turn off the equipment power supply or remove the wall socket directly, and cut off the water at the same time. Check whether the insurance tube in the control system is melted, just replace it.
If you cannot judge the cause of the accident, please contact the company in order to get the emergency maintenance. 7. When the constant temperature and humidity display cabinet is installed for the first time or when it is closed for a long time, the system must have a cycle of about 24 hours.
After several work circulation.